Non-clinical testing has demonstrated that the UroLift Implant is MR Conditional. A patient with this device can be safely
scanned in an MR system immediately after placement meeting the following conditions1:
Under the scan conditions defined above, the UroLift Implant is expected to produce a maximum temperature rise of 2.4°C after 15 minutes of continuous scanning (i.e., per pulse sequence)
In non-clinical testing, the image artifact caused by the device extends approximately 15 mm from the UroLift Implant when imaged with a gradient echo pulse sequence and a 3.0 Tesla MRI system.1
The safety of the delivery system has not been evaluated in the MR environment, and therefore, the delivery system should
not be used within the MR environment.
If you have any questions about MRI safety, please consult your doctor.